True Color Analysis of Transmissive Liquids
Color analysis is any technique by which an unknown color is evaluated in terms of known colors. Color analysis is widely used in scientific studies involving the appearance of objects and lights, and is of great importance for the analysis of raw materials and finished products of industry. Color Analysis is also important for monitoring process liquids and as a measurement of water quality for distribution or discharge. Color in water or wastewater may result from the presence of metals, organic acids, microbiological matter and/or industrial wastes.
Transmissive color analysis is characterized by the technique of measuring light that has passed through a partially opaque sample (as opposed to reflective analysis which measures light that is reflected from the surface of a solid sample).
Color in partially opaque or transparent liquids is determined by the process of measuring and comparing light transmitted through a sample with light that has been transmitted through a color standard, such as a glass disk or an opaque standard solution.
Apparent vs True Color
In a liquid, turbidity in the form of colloidal matter or suspended particulates can scatter the light transmitted through the liquid in a way that alters the observed color. If a color measurement is made without first removing the colloidal or suspended sources of turbidity, the resulting measurement is known as apparent color. True color is the measurement made following the removal of colloidal or suspended sources of turbidity. The color of liquids is also a pH dependent, increasing in color with increased pH, which may require pH measurement or buffering.
There are a variety of techniques used by laboratories to remove turbidity prior to sample analysis, including filtration through filter paper, centrifugation and other techniques. Some of these techniques were originally developed to extract solids for analysis. If these techniques are used for color analysis sample preparation, they may actually alter true color in the sample by exposing the sample to dyes and bleaches used to process the filter media.
Pure white light such as sunlight actually contains all of the colors of the spectrum. Each component color is refracted differently when passing from one medium into another medium with a different refractive index. This is why sunlight is separated into the color spectrum when passing through a prism. A complimentary refraction occurs if the light passes back to the original medium.
Transmitted vs Reflected Light
Transmissive colored substances are neither fully transparent (transmitting all colors of light) or fully opaque (transmitting none of them). Instead, transmissive colored substances are transparent to some colors and opaque to others. The object passes or reflects some colors and absorbs others, deriving the observed (by the eye) color from the fractions that have not been absorbed. Thus, a relatively transparent liquid will absorb some components of white light, while allowing other components to pass through. Detection is performed directly through the liquid, with the light source and detector arranged in a straight line. A relatively opaque or solid object with color will also absorb some fraction of white light, while reflecting other fractions. Refractive measurement is made at an angle, with different results obtained at different angles. ChemScan is designed for the analysis of transmissive liquids and relies on the percent of light transmission at numerous wavelengths for analysis.
Physical vs Physiological color
Light that is colored by transmission through a colored media or reflected from a colored surface may not contain the exact physical color corresponding to the physiological color that we observe with our eyes. Physical color can be measured by one of the techniques discussed below. The eye detects color through three types of color sensitive retinal cells, reacting to red, green and blue, with the brain combining these individual sensations to interpret color. An object that consists of two individual physical colors (such as red and yellow) will be perceived physiologically to be a combined color (orange). Therefore, color measured by physical means may show the responses from individual color components, while the eye will see the physiological results from the combination of the individual color components.
Color Measurement Terminology
The true color of a transmissive sample is the color measured in the sample after the turbidity from suspended or colloidal matter has been removed through filtration or other liquid-solid separation processes. We refer to this as a filtered sample.
The color of a filtered sample can be expressed in terms of three values: The hue is designated by the term “dominant wavelength,” the degree of brightness is designated by the term “luminance,” and the saturation (pale, pastel, etc.) is designated by the term “purity.”
Color Measurement Methods
Platinum-Cobalt (Pt-Co) Methods
The tristimulus filter method (Standard Method 2120D) is based on the measurement of percent transmittance at three specific wavelengths (590, 540, and 438nm). Luminance plus three tristimulus values are calculated directly from percentage transmittance values at the three wavelengths. The ChemScan Process Analyzer can directly detect these measurements and apply a special analysis algorithm to identify points on the chromaticity diagram. Results can be reported in terms of dominant wavelength (hue wavelength), percent luminance and percent purity. Or values can be converted to an ADMI single number color difference (EPA method 110.1 and Standard Method 2120E) using the Adams-Nickerson Color Difference (DE) calculation. Differences between sample measurements and standards can also be made using APHA color values.
ASA Analytics, 2325 Parklawn Drive, Suite I, Waukesha, WI 53186 PH 262-717-9500,